Il quoziente di stabilità implantare (ISQ) è un oggettiva unità di misura della stabilità implantare. Il range clinico dell’ISQ è normalmente tra 55 e 80. I valori più alti sono normalmente registrati nella mandibola rispetto alla mascella. La scala ISQ non ha una correlazione lineare con la micro mobilità. Con più di 1000 studi scientifici, adesso sappiamo che una buona stabilità deve avere un indice ISQ di almeno 70, un valore tra il 60 e il 69 indica una stabilità media e sotto i 60 una stabilità bassa.
Se il valore di ISQ iniziale è alto, una piccola diminuzione della stabilità è fisiologica e normalmente si assesta con il tempo. Una grossa diminuzione della stabilità od una continua diminuzione devono essere considerati come segnali d’allarme.
I valori più bassi dovrebbero invece aumentare dopo il periodo di guarigione, una diminuzione in questo caso potrebbe invece indicare un insuccesso dell’impianto e richiederebbe un azione correttiva.
1. 20 Jahre Erfahrung mit der Resonanzfrequenzanalyse
Sennerby L Prof, Sahlgrenska Academy, University of Gothenburg, Sweden
Implantologie 2013;21(1):21-33 (In German)
Translated from German “It is likely that ISQ measurements can be used as one additional parameter for diagnosis of implant stability and decision-making during implant treatment and follow-up. The threshold values are the present author’s own somewhat conservative suggestions based on own experience and other values may be relevant for other clinicians and implant designs. The green zone contains “safe” implants showing primary ISQ values from, for instance 70 and above. The red zone contains “questionable” implants with an ISQ value below for instance 55. The yellow zone represents implants with an ISQ from 55 to 70”.
2. Immediate vs. early loading of SLA implants in the posterior mandible: 5-year results of randomized controlled clinical trial.
Kokovic V, Jung R, Feloutzis A, Todorovic V, Jurisic M, Hämmerle C
Clinical Oral Implants Research, 00, 2013, 1-6
After 5 years, survival in the both groups was 100 %. The mean value of primary implant stability was 76,92 ± 0,79 ISQ. In the first 6 weeks, ISQ values significantly increased in the test group as well as in the control group. Based on these results, the self-tapping implants inserted in posterior mandible can provide adequate primary stability value as the main factor for immediate and early loading protocol.
3. Early Loading of Nonsubmerged Titanium Implants with a Chemically Modified Sand-Blasted and Acid-Etched Surface: 6-Month Results of a Prospective Case Series Study in the Posterior Mandible Focusing on Peri-Implant Crestal Bone Changes and Implant Stability Quotient (ISQ) Values
Michael M. Bornstein, Dr. med. dent.; Christopher N. Hart, DMD; Sandro A. Halbritter, Dr. med. dent.; Dean Morton, BDS, MS;† Daniel Buser, Prof. Dr. med. dent.
Clin Implant Dent Relat Res 2009
If the ISQ value at day to load is < 65, an additional healing period is recommended, and the ISQ values is measured again 3 weeks later until the required level is reached. This approach is practical and well understood by patients. (Prof. Daniel Buser prefers ≥ 70 ISQ, single teeth, early loading/Straumann, otherwise add three weeks, according to an oral presentation given at the Osstell Scientific Symposium in connection to the of the EAO 2010.)
4. The Predictive Value of Resonance Frequency Analysis in the Surgical Placement and Loading of Endosseus Implants
Baltayan, Serge; Mardirosian, Martin; El-Ghareeb, Moustafa; Aghaloo, Tara; Pi-Anfruns, Joan; Moy, Peter
AAID Poster 2011
One-stage placement of implants with ISQ values greater than 66 can be performed. Implants with ISQ values less than or equal to 66 should be placed using the two-stage protocol, which shows a higher survival rate. The computed ISQ = 66 cut-off value used to select between one-stage and two-stage placement is validated in this study. Moreover, early loading of implants with ISQ values greater than 64 can be performed. Implants with ISQ values less than 64 should utilize traditional loading, which shows a higher survival rate. The computed ISQ = 64 cut-off value used to select between early and traditional loading is validated in this study. Higher ISQ values at osseointegration correlate with higher survival rates.
5. Direct Loading of Implants
Pär-Olov Östman DDS, PhD, MD, Private practitioner, Falun- and Biomaterial group, Sahlgrenska
Academy, Gothenburg – Tandläkartidningen årg 100 nr 3, 2008
20 consecutive patients with totally edentulous maxillas were included in the study. The criteria for direct loading was insertion torque 30 Ncm and an ISQ > 60 on the most posterior implants and a sum of 200 ISQ (average 50 ISQ) on the 4 anterior
implants. The overall conclusion with the thesis is that dental implants can be direct loaded with a good result if high primary stability can be obtained and if a stable provisional bridge with good occlusion is splinting the implants.
6. Diagnosis of Implant Stability and its Impact on Implant Survival: A Prospective Case Series Study
Daniel Rodrigo, Luis Aracil, Conchita Martin, Mariano Sanz
Clin. Oral Impl. Res. 21, 2010; 255-261
The evaluation of RFA values to assess implant secondary stability (Osstell 2) demonstrated a
statistically significant correlation with implant outcome. In fact, no implant with ISQ > 60 failed, while 19 % of implants with ISQ < 60 failed.
7. The relationship between resonance frequency analysis (RFA) and lateral displacement of dental implants: An in vitro study
Pagliani L, Sennerby L, Petersson A, Verrocchi D, Volpe S & Andersson P
Journal of Oral Rehabilitation 2012
Both RFA and displacement measurements correlated with bone density. It is concluded that RFA measurements reflect the micromobility of dental implants, which in turn is determined by the bone density at the implant site. The correlation between ISQ and micron was non-linear and micro motion was reduced with app. 50 % from 60 ISQ to 70 ISQ.
8. Implant Stability Quotient (ISQ) vs Direct in Vitro Measurement of Primary Stability (Micromotion): Effect of Bone Density and Insertion Torque
Paolo Trisi PhD, Teocrito Carlesi DDS, Marco Colagiovanni DDS, Giorgio Perfetti MD, DDS
Journal of Osteology and Biomaterials, Volume 1, Number 3, 2010
Results showed a high dependence between the observed micromotion and the ISQ values, indicating that micromotion decreased with increasing ISQ values. An in vitro study and the results cannot be directly transferred to clinical applications.
9. Determination of Primary Stability: A Comparison of the Surgeon’s Perception and Objective Measurements
Degidi M, Daprile G, Piattelli A. – Int J Oral Maxillofac Implants. 2010 May-Jun;25(3):558-61.
The accuracy of primary stability prediction is not good enough to prevent mistakes when using an immediate loading technique, thus a more systematic use of objective measurements has to be encouraged.
– Low perception of RFA-value was correct in 19 % (81 % was either medium or high RFA-value)
– Medium perception of RFA-value was correct in 43 % (57 %was either low or high)
– High perception of RFA-value was correct in 85 % (15 % was either low or medium)
10. Implant Stability Measurements Using Resonance Frequency Analysis: Biological and Biomechanical Aspects and Clinical Implications
Sennerby L, Meredith N.
Periodontology 2000, Vol. 47, 2008, 51–66.
This article summarizes much of the science around Osstell and ISQ. It describes what implant stability is and how it can be measured. It talks about clinical implications of measuring ISQ, as well as how different ISQ levels and ISQ trends can be interpreted.